To correctly compute the active power in three phase panels, each current sensor must be set with the matching phase line which powers the current sensor wire.
The Phases 1, 2 and 3 respectively match the voltage inputs L1,L2 and L3.
The x phase to be set is the one for which the voltage between the wire and the concentrator's Lx input is null (the others are around 400 Volts).
In the case of a mono phase panel, all channels must be set to the only phase (usually L1) which powers the concentrator.
To not forget to configure a phase, you can use the initialization function on the touchscreen module (cf next video)
Initialize a Multivoies system:
Set recording period:
With Multivoie system, data are continuously recorded. A period change creates a new line in the data file with the time the change occurred.
Download datas to SD card:
Before using GPRS module, you must create the correct folder on the ftp server
\"FTP folder set in GPRS module"\"Concentrator SN"\ (example : \0002\01001224\)
The GPRS module list must be up to date (button UPD. LIST)